Approved DAMAS for North America
Pd-based, Type 4, metal-ceramic dental alloy.
PLEASE READ BEFORE MAKING ANY MODIFICATIONS TO THE FRAMEWORK!
ONLY AREAS IN NEED OF ADJUSTING NEED METAL FINISHING. The entire surface does not require refinishing.
Perform desired finishing with dedicated carbide burs or non-contaminating stones such as grinding instruments categorized as non-contaminating, having ceramic binderss. ALL OTHER SURFACES CAN BE SANDBLASTED. IT IS NOT NECESSARY TO PERFORM TRADITIONAL METAL FINISHING ACROSS THE ENTIRE SURFACE.
When finishing steps are completed, sandblast the entire surface using non-recycled 50–100-micron AL2O3 at a maximum of 75 psi pressure. Using steam or ultrasonic methods, clean the framework with distilled water or ethanol and allow it to dry.
Clean using distilled H2O in an ultrasonic, or steam clean and dry the part free from any water residue. Place the framework on firing trays providing stable support. Do not touch the dried surface. Begin firing at 540C/1000F, increasing the temperature to 950C/1740F without vacuum with a 1-minute hold at the high temperature.
After glazing remove oxide and flux residue and finish/polish the framework with rubber finishers/polishers.
Inlays, Onlays, 3/4 Crowns, Crowns, Telescope Crowns, Conus Crowns, Posts-Cores, Bars, Short and Long Span Bridges, Bars, Implant Retained Superstructures, Partial Dentures.
Consultation with a physician is recommended for patients with known allergy/sensitivity to any major or minor constituents of this alloy. This material is not recommended for any application not within the indications listed.
In individual cases, sensitivity, or allergies to components of this alloy may occur. Strategy Milling makes no claims regarding the MRI compatibility of its dental alloys. The patient must be made aware of the possibility that dental alloys may affect MRI results and disclose the presence of dental alloys to the MRI Technician before tests are performed.
Galvanic effects may occur between different alloys in the same oral environment.
Vapors and metal dust are harmful if inhaled. Please use extraction equipment and PPE when working with this material.
This material has been developed solely for use in dentistry. Processing should be carried out strictly according to the Instructions for Use. Strategy Milling cannot be held liable or accept responsibility for damages resulting from failure to observe the Instructions or the stipulated application area. The user is responsible for testing the products for suitability and use for any purpose not explicitly stated in the Instructions. These regulations also apply if the materials are used in conjunction with other manufacturers' products.